The aging chamber is used for testing various performance indexes of components and materials of electronic components, automobile and motorcycle, rubber, plastic, metal, weapon, ships, universities and scientific research units and other related products under the changes of constant temperature and high temperature. For this kind of precise instrument, we need to know more to prolong the service life of aging chamber, but do you know what needs to be noticed to prolong the life of aging chamber? If you want to extend the lifespan of aging chamber, you can start from the following three aspects:
Check the insulation of the power circuit and no leakage before switching on the power. There shall be no contact between the resistance wires of the heater to prevent short circuit.
Open the exhaust valve before electricity.
Use non-explosion-proof aging chamber and do not allow volatile items to enter the aging chamber.
Observe and adjust the temperature in the aging chamber any time, which should meet the requirements of the drying process.
Keep the electric chamber clean, check and clear the oxide scale beside the resistance wire in the chamber.
Cut off the power and close the aging chamber.
The equipment should be operated by a special person. Non-professional maintenance personnel are forbidden to disassemble or adjust the internal electricity parameters of the equipment.
No inflammable and explosive materials in the equipment display area but place effective fire-fighting facilities.
The equipment should be placed on a flat and firm ground and the placing area should be well ventilated with special exhaust channel.
Prepare the power line to fit equipment power and make up the circuit breaker within 2 meters of the equipment.
Take effective protective measures to prevent scalding when taking product from the equipment.
High temperature chamber belongs to normal temperature oven: no baking flammable and explosive articles.
Connect the earth wire safely and correctly.
Operators must strictly follow the operating rules and shall not violate the operation in rush hours.
Always keep the appearance of the machine clean and tidy; Cover the chamber with plastic dust cover and place in dry room if not used for a long time.
Regularly check whether the turntable, motor and reducer are in normal condition.
Check whether the cooling fan operates normally and add lubricating oil.
Regularly check whether the circuit breaker, overload and overtemperature protector are normal.
The above is whatshares with you. Please contact us if you want to know more about aging chamber.