walk-in chamber After long-term use, some components or systems may have certain problems, such as the water system of the walk-in chamber equipment may have sediment and other issues. In order to avoid affecting the detection effect of the walk-in chamber equipment, it is necessary to regularly clean the water system of the walk-in chamber equipment. What is the specific process of cleaning operation?
In fact, the cleaning work of the equipment water system is periodic, and everyone should persistently clean the water system of the walk-in chamber (usually cleaned once every three to four months). The main process of cleaning the walk-in chamber equipment is:
First, open the equipment room door and turn off the total power supply of the equipment downwards. Rotate the drain valve of the system to the open position. At this time, the water is returned to the sewage tank through the return pipe and all the water is drained into the sewer. Pull out the return pipe, and unplug the power connector of the pump motor and the outlet pipe of the pump motor.
If there is leakage at the outlet of the pump motor of the walk-in chamber equipment at this time, don't panic. This is a normal phenomenon. Just hold the outlet of the pump motor with your hand, and quickly drain the water in the sewage. After cleaning the various components of the system, put the sewage bucket back in place, and connect the power cord and connector of the pump motor outflow pipe, and pull the outflow pipe of the pump motor back. Turn over the cover of the sewage bucket and pour distilled water or pure water into it. Then rotate the drain valve to close it, and turn on the total power supply of the test equipment. At this time, the water will be drawn from the sewage bucket and the pump motor to various components of the water system, and the cleaning work of the water system is officially completed.
The water system of the walk-in chamber is an important factor affecting its humidity. Many people often do not pay attention to the maintenance of the equipment after purchasing it, which may affect the accuracy of the equipment if it is not cleaned for a long time, and even more seriously cause equipment failure and interrupt tests. Therefore, everyone should value the cleaning work of the equipment in daily life. If any parts are found to be aging or damaged during cleaning, they need to be replaced in time.